Gas meter METRIX G 2,5
Household diaphragm gas meter Metrix UG2,5 is intended for commercial accounting of the volume of gas consumed, primarily in living conditions, in which the total-maximum use of gas with installed gas equipment does not exceed 4 m3/h.
Gas meters Metrix UG2,5 are used for commercial accounting of natural gas, gaseous propane, butane or their mixtures.
On request for remote information transmission, meters can be made with a low-frequency pulse sensor (giron) type NI-3.
The pulse sensor can be connected to a meter already installed by the customer. The magnitude 1 of the pulse is 0.01 m3.
Type UG 2,5
Maximum gas flow rate - Qmax m3/h 4
Minimal gas flow rate - Qmin m3/h 0.025
Nominal gas flow - Qnom m3/h 2.5
Cyclic volume dm3 1.3
Maximum working pressure - Pmax kPa 50
The maximum value of the adder is 99999.999 m 3
Limits of the relative error of the meter in the range of costs, at the initial calibration
- Qmin to 0.1 Qnom - E
- 0.1 Qnom to Qmax - E
Interval distance%% mm ± 3 ± 1,5 110
Sensitivity threshold dm3 / h 5
Pressure loss Pa ≤ 200
Range of ambient temperatures ° C -40 ... +55
Operating temperature range ° C -25 ... +55
Threaded fittings in. ¾ ", 1" 1/4
How to order?
If you want to buy a gas meter on our site, please click the "Order" button. On this page, fill out a special form with your name, phone number and basic details of the order. If necessary, you can call us on the specified phones.
Delivery of the goods is carried out by the territory of Ukraine by carriers "Nova Poshta" or "Delivery".